Fisher Isotemp 637G oven
Inlet air enters through a port located under the oven floor. Heatgenerated convection then gently moves the air in a vertical circulation pattern. Exhaust air is vented through a port at the oven top. A safety backup is built into the controller: if the primary heater control fails, the backup will maintain control at 5 o C above the set point. An alarm LED then indicates that the backup controller is operating the oven. A circuit breaker protects the oven from power surges.
All of our equipment is used and reconditioned. Our technicians test ovens using a calibrated digital chart recorder to verify the accuracy of temperature controls and readouts. We replace any hardware or parts as is necessary and clean all incubators thoroughly prior to shipment.
Fisher Isotemp 637G ovens specifications
Operating Range: 50°C to 225°C
Average Uniformity @ 200°C: ± 4.5 o C
Control Sensitivity: ± 0.5°C
Recovery time @ 200°C (door opened one minute): 4.0 minutes
Rise Time to 225°C: 60 minutes
Chamber Volume: 3.8 cu ft.
Chamber Dimensions (W x D x H): 18 x 18 x 20
American Instrument Exchange, Inc.
1023 Western Avenue
Haverhill, MA 01832