4E's USA - Micro-Volume Spectrophotometer - Nano-3100

2024-08-05T16:30:45-04:00

Vendor: 4E's USA
Type: Spectrophotometers/Colorimeters
Price: 0.00

4E's USA Micro Volume Spectrophotometer Nano-3100 - NEW

The 4E's USA Micro Volume Spectrophotometer Nano-3100 measures the absorbance of entire wavelength (185-910nm). In pedestal measurement, the instrument can measure 0.5-2μL samples with high accuracy and reproducibility. The sample retention system employs surface tension to hold the sample in the place between two optical fibers. This enables the measurement of very highly concentrated samples without the need for dilutions.

Specifications

Model: Nano-3100

Wavelength range: 185-910 nm

UV-Vis

Pedestal measurements

Manufacturer’s warranty

Features

The 4E’s USA Nano-3100 with built-in 7-inch color touch screen, can accomplish all detection functions without connecting to a computer. The device comes with an intuitive user interface, and the compact design saves valuable laboratory space. The device has wide application in molecular biological experiments, such as DNA, RNA, protein detection.

Using this technology, Micro-volume Spectrophotometer has the capability to measure sample concentration up to 200 times higher than the samples measured by a common spectrophotometer. Height of liquid between optical fibers can be adjusted in real-time way to get more precise data. When powered on, it runs the self-testing. There is no need to preheat before measuring. All operations are simple and convenient.

Wide detection range: With a 0.03 mm minimum detection path length, the maximum detection concentration of the device is as high as 27,500 ng/μL, and the minimum detection concentration as low as 0.2 ng/μL dsDNA.

Stable and reliable detection value: Stable and Accurate detection for both low and ultra-high concentrations.

Large size color touch screen: A large 7-inch high-resolution color touch screen with intuitive UI enables to easily perform detection and analyze results.

Indication Lights: A lighting strip around the test area flashes to indicate different test stages.

4E's USA Nano-3100 - Additional Specifications

Detector | 2048-element CCD linear image sensor
Light Source | Xenon flash lamp
Minimum Sample Volume (uL) | 0.5
Pathlength (mm) | 0.03/0.05/0.1/0.2/1.0 auto-ranging
Wavelength Range (nm) | 185-910
Wavelength Accuracy (nm) | ±1
Spectral Bandwidth | ≤1.8 (FWHM at Hg 253.7nm)
Photometric (Abs) | 0-550 (10mm equivalent)
Measurement Repeatability | 0.002 (1mm path) or 1% CV
Photometric Accuracy | ±2% (at 0.86A at 257mm) 3% (at 0.64A at 350mm)
Limit of Detection | Pedestal: 2ng/uL dsDNA, 0.06mg/mL BSA，0.03mg/mL IgG
Maximum Concentration | Pedestal: 27500ng/uL dsDNA, 820mg/mL BS，400mg/mL lgG
Measurement Time(S) | ≤5

Pictures

The photos above are those found on 4E's USA's website, unless we’ve had one in stock. At that point, we take our own pictures of the equipment for our website before sending them out to the customer. We will provide photos of your actual equipment once it arrives at your request.

For inquiries about the 4E's USA Micro Volume Spectrophotometer (UV-VIS) 3100, please give us a call at 978-521-2221, or fill out the form on this page.