4E's USA – Micro Volume Spectrophotometer - NEW

2024-10-16T15:49:45-04:00

Vendor: 4E's USA
Type: Spectrophotometers/Colorimeters
Price: 8600.00

4E's USA – Micro Volume Spectrophotometer (UV-VIS)

The 4E’s USA Micro Volume Spectrophotometer is used in the detection of micro-volume quantities of nucleic acids, proteins, chemical substances and liquids in laboratory, food industries, chemistry, microbiology, medicine research and development.

4E's USA – Micro Volume Spectrophotometer (UV-VIS) - Specifications

Model: SPT-Nano

Wavelength range: 190 - 800 nm

Pedestal and cuvette measurements

Manufacturer’s warranty

4E's USA Micro Volume Spectrophotometer (UV-VIS) – Features

The Micro volume spectrophotometer (UV-Vis) with a built-in 7-inch color touch screen can perform all detection functions without connecting to a computer. The device has a full spectral range (190-800nm) for accurate quantitative analysis of DNA, proteins, etc. It comes with an intuitive user interface for easy operation.

The instrument requires only 0.5-2μl micro-volume samples for accurate and reproducible measurements and can detect high concentration samples without dilution. The Xenon flash lamp light source ensures that the device is ready for immediate use without any warm-up time and provides high stability and a long operating life.

The Micro Volume Spectrophotometer offers three different modes of operation: Microvolume Measurement and OD600nm Measurement. In the microvolume mode, it requires only 0.5 - 2μl micro-volume samples for accurate and reproducible measurements. Additionally, the OD600 mode is specifically designed for measuring the optical density of bacterial cultures, providing valuable information for cell growth and viability studies.

4E's USA Micro Volume Spectrophotometer (UV-VIS) – Highlights

7-inch LCD touchscreen with intuitive user interface accomplishes detection without the need to connect to a computer

Only requires 0.5 - 2µl sample volume for accurate measurement.

Full spectral range of wavelength detection from 190nm to 800nm.

Simply power it on and start measuring, no need for lamp warm-up.

The Xenon flash lamp has long service life up to 10 years.

Automatic detection on lowering the pedestal arm.

OD600 function to detect concentration of bacteria and microorganisms.

The test result can be printed by a built-in printer or exported via USB for data analysis and storage.

4E's USA Micro Volume Spectrophotometer (UV-VIS) – Additional Specifications

Model | SPT-NanoF (with built-in Fluorescence Detection)
Wavelength range | 190-800nm
Minimum sample size | 0.5～2μL
Path length (auto-ranging) | 0.05mm (high concentration); 0.2mm (high concentration); 1mm (normal concentration)
Light source | Xenon flash lamp
Detector type | 2048px, CMOS
Wavelength accuracy | 1nm
Spectral resolution | ≤3nm
Absorbance precision | 0.003Abs
Absorbance Accuracy | 1% (4.096A at 260nm)
Absorbance Range | 0.04～300A
Lower Limit of Detection | 2 ng/μL(dsDNA); 0.06 mg/mL (BSA)
Maximum Concentration | 15000 ng/μL(dsDNA); 440 mg/mL (BSA)
Sample pedestal material | Aluminum alloy and Quartz fiber
Power supply | AC100V-240V 50Hz/60Hz (power adapter)
Operating power | 48W
Standby power | 8W
Software | Android
Data Output | USB 2.0, built-in printer
OD600nm Measurement
Sample Holder Type | Cuvette
Cuvette Dimensions | 12.5x12.5x45mm; Z-Height: 8.5mm
Light source | LED
Wavelength range | 600±8nm
Absorbance range | 0～4A
Dimensions and Weight
Dimension (W x D x H) | 318 x 220 x 205mm
Weight | 4.1kg

4E's USA Micro Volume Spectrophotometer (UV-VIS) – Pictures

Since we often stock multiple units of each model, we may not feature photos of every unit. However, if you request, we will provide photos of your specific equipment after we test and clean it.

For inquiries about the 4E's USA Micro Volume Spectrophotometer, call us at 978-521-2221 or fill out the form on this page.