4E's USA - Micro-Volume Spectrophotometer - Nano-2100

2024-08-05T16:30:47-04:00

Vendor: 4E's USA
Type: Spectrophotometers/Colorimeters
Price: 0.00

4E's USA – Micro Volume Spectrophotometer Nano-2100 - NEW

The 4E’s USA Micro Volume Spectrophotometer Nano-2100 features a built-in 7-inch color touchscreen, allowing users to perform all detection functions without needing a computer. Its intuitive user interface simplifies operation, while its compact design conserves valuable laboratory space. This advanced device plays a crucial role in molecular biological experiments, including DNA, RNA, and protein detection, making it a versatile tool for researchers.

Designed for precise measurements, the Nano-2100 is a micro-volume UV-Vis spectrophotometer that determines absorbance at fixed wavelengths. It supports pedestal measurement, enabling the NanoReady Touch to accurately measure 0.5-2μL samples with high reproducibility. This capability ensures reliable results even when working with small sample volumes, making the device ideal for laboratories requiring precision and efficiency.

Specifications

Model: Nano-2100

Wavelength range: 185-910 nm

UV-VIs

Pedestal measurements

Manufacturer’s warranty

Features

The NanoReady Touch utilizes an advanced sample retention system that employs surface tension to hold the sample securely between two optical fibers. This innovative design enables the measurement of highly concentrated samples without requiring dilution.

With this technology, the NanoReady Touch can measure sample concentrations up to 200 times higher than those analyzed by a common spectrophotometer. Additionally, users can adjust the liquid height between optical fibers in real time to achieve more precise data.

Enhancing user convenience, the device automatically runs a self-test upon powering on, ensuring optimal functionality before each measurement. Unlike many spectrophotometers, it does not require preheating, allowing users to begin testing immediately. This feature saves time and enhances workflow efficiency in the laboratory.

The 4E’s USA Micro Volume Spectrophotometer 2100 offers a wide detection range with a minimum path length of 0.03 mm. It accurately detects concentrations as high as 27,500 ng/μL and as low as 0.2 ng/μL dsDNA, ensuring precision for both high and low sample concentrations.

Designed for stability and reliability, the device provides accurate detection across a broad concentration range. The large 7-inch high-resolution color touchscreen offers an intuitive interface, making it easy to perform tests and analyze results. Additionally, an integrated lighting strip around the test area flashes to indicate different test stages, enhancing user awareness and efficiency.

4E's USA Nano-2100 – Additional Specifications

Detector | 2048-element CCD linear image sensor
Light Source | Xenon flash lamp
Minimum Sample Volume (uL) | 0.5
Pathlength (mm) | 0.03/0.05/0.1/0.2/1.0 auto-ranging
Wavelength Range (nm) | Default 230/260/280/350/600/185-910 customizable
Wavelength Accuracy (nm) | ±1
Spectral Bandwidth | ≤1.8 (FWHM at Hg 253.7nm)
Photometric (Abs) | 0-550 (10mm equivalent)
Measurement Repeatability | 0.002 (1mm path) or 1% CV
Photometric Accuracy | ±2% (at 0.86A at 257mm) 3% (at 0.64A at 350mm)
Limit of Detection | Pedestal: 2ng/uL dsDNA, 0.06mg/mL BSA，0.03mg/mL IgG
Maximum Concentration | Pedestal: 27500ng/uL dsDNA, 820mg/mL BS，400mg/mL lgG
Measurement Time(S) | ≤5

Pictures

The photos above are those found on 4E's USA's website, unless we’ve had one in stock. At that point, we take our own pictures of the equipment for our website before sending them out to the customer. We will provide photos of your actual equipment once it arrives at your request.

For inquiries about the 4E's USA Micro Volume Spectrophotometer Nano-2100, please give us a call at 978-521-2221, or fill out the form on this page.