4E's USA - Micro-Volume Spectrophotometer - NanoC-2100

2024-08-05T16:30:46-04:00

Vendor: 4E's USA
Type: Spectrophotometers/Colorimeters
Price: 0.00

4E's USA – Micro Volume Spectrophotometer (UV-VIS) C-2100 - NEW

The 4E’s USA Micro Volume Spectrophotometer NanoC-2100 features a built-in 7-inch color touchscreen, allowing users to perform all detection functions without connecting to a computer. With an intuitive user interface, the device ensures ease of use, while its compact design saves valuable laboratory space. It has broad applications in molecular biological experiments, including DNA, RNA, and protein detection.

Specifications

Model: NanoC-2100

Wavelength range: 185-910 nm

UV-Vis

Pedestal and cuvette measurements

Manufacturer’s warranty

Features

The NanoC-2100 is a micro-volume UV-Vis spectrophotometer designed to measure absorbance at fixed wavelengths. It supports both pedestal measurement for small-volume samples and traditional cuvette measurement. In pedestal mode, the NanoReady Touch can accurately measure 0.5-2μL samples with high reproducibility.

The sample retention system uses surface tension to hold the sample between two optical fibers, eliminating the need for dilutions when measuring highly concentrated samples. This advanced technology allows the NanoReady Touch to measure sample concentrations up to 200 times higher than those analyzed by standard spectrophotometers. Additionally, the liquid height between optical fibers can be adjusted in real time to obtain more precise data.

For samples with low concentration, high volatility, or low surface tension that cannot form a liquid bridge, cuvette measurement is recommended. When powered on, the device runs a self-test automatically, eliminating the need for preheating before measurement.

The NanoC-2100 offers a wide detection range, with a minimum path length of 0.03 mm, allowing it to detect concentrations as high as 27,500 ng/μL and as low as 0.2 ng/μL dsDNA. It delivers stable and accurate results for both low and ultra-high concentrations.

The large 7-inch high-resolution color touchscreen provides an intuitive interface for easy detection and analysis. Additionally, an indicator light strip around the test area flashes to signal different test stages, enhancing user convenience.

4E's USA C-2100 – Additional Specifications

Detector | 2048-element CCD linear image sensor
Light Source | Xenon flash lamp
Minimum Sample Volume (uL) | 0.5
Pathlength (mm) | 0.03/0.05/0.1/0.2/1.0 auto-ranging
Wavelength Range (nm) | Default 230/260/280/350/600/185-910 customizable
Wavelength Accuracy (nm) | ±1
Spectral Bandwidth | ≤1.8 (FWHM at Hg 253.7nm)
Photometric (Abs) | 0-550 (10mm equivalent)
Measurement Repeatability | 0.002 (1mm path) or 1% CV
Photometric Accuracy | ±2% (at 0.86A at 257mm) 3% (at 0.64A at 350mm)
Limit of Detection | Pedestal: 2ng/uL dsDNA, 0.06mg/mL BSA，0.03mg/mL IgG
Maximum Concentration | Pedestal: 27500ng/uL dsDNA, 820mg/mL BS，400mg/mL lgG
Measurement Time(S) | ≤5
Cuvette Temperature (℃) | 37±0.5
Cuvette Stirring Speed | 10-900 RPM 10 levels (adjustable)
Cuvette Photometric Range | 0-1.5A (10mm)
Cuvette Limit of Detection | 0.2ng/μL dsDNA, 0.006mg/mL BSA, 0.003mg/mL lgG

Pictures

The photos above are those found on 4E's USA's website, unless we’ve had one in stock. At that point, we take our own pictures of the equipment for our website before sending them out to the customer. We will provide photos of your actual equipment once it arrives at your request.

For inquiries about the 4E’s USA Micro Volume Spectrophotometer NanoC-2100, please give us a call at 978-521-2221, or fill out the form on this page.