Thermo 3307 SteriCult incubator

2023-01-24T08:49:34-05:00

Vendor: Thermo
Type: Incubators
Price: 5200.00

Thermo 3307 SteriCult incubator

The Thermo 3307 SteriCult incubator is ideal for cell culture work. Featuring integral 140°C high-temperature sterilization function, and optimized airflow system that will not interfere with samples or incubator function.

Thermo 3307 SteriCult incubator features

Single 8.2 cubic foot chamber

Temperature range: ambient +5°C to 50°C

HEPA air filter provides Class 100 air quality

Upgraded to new CO2 sensor available

90-day warranty

In-chamber HEPA air filtration continuously removes particulates for Class 100 quality conditions, the Thermo 3307 allows users to carry out high-volume culturing with strict contamination controls. The on-demand 140°C sterilization cycle eliminates contamination from internal surfaces and an audible alarm activates if outer door is opened during the cycle or if temperature reaches 60°C (140°F) or greater. An access code prevents accidental initiation of the cycle.

We've purchased and installed a new, upgraded infrared CO2 sensor for these 3307's, a $3,000 value.

Testing

Our technicians test CO2 incubators using a calibrated digital chart recorder to verify the accuracy of temperature controls and display, as well as using a Fyrite indicator to measure CO2 levels in the chamber. We replace any hardware or parts as is necessary and clean all incubators thoroughly prior to shipment.

Click Here to see the Thermo 3307 on the American Instrument Exchange YouTube channel.

Thermo 3307 incubator specifications

Temperature

Control - ±0.1°C @ 37°C (98.6F)

Range - 5°C over ambient to 50°C (122F)

Sensor - Thermistor

Controller - Microprocessor

Setpoint - Digital

Display - Digital LED

Readability & Setability - 0.1°C

Uniformity - ±0.2°C @ 37°C (98.6F)

Temperature Safety

Sensor - Thermistor

Controller - Microprocessor

Setability - 0.1°C

CO2

Control - ±0.1% @ 5.0%

Range - 0-20%

Calibration - Auto-zero

Inlet Pressure - 15 PSIG (1.0 bar)

Filter - 0.2 micron, disposable

Sensor - Dual beam IR

Controller - Microprocessor

Display - Digital LED

Readability & Setability - 0.1%

Tracking Alarm - User programmable high/low

Controlled RH

RH - Ambient to 95% RH, non-condensing

Humidity Control - ±2.0%

Sensor - Capacitive

Controller - Microprocessor

Readability & Setability - 1%

Humidity Reservoir - 1.0 gal. (3.8 liters)

Tracking Alarm - User programmable high/low