Thermo Heracell VIOS 250i LK CO2 incubator

2025-12-30T15:31:59-05:00

Vendor: Thermo Scientific
Type: Incubators
Price: 5200.00

Thermo HERAcell VIOS 250i LK CO2 Incubator

The Thermo Heracell VIOS 250i LK CO2 incubator is specifically designed to deliver the performance reliability, ease of operation, and value required to support a range of culturing needs from basic research to the demanding, leading-edge applications, so you're ready for whatever comes next.

Thermo Heracell VIOS 250i LK CO2 Incubator Specifications

Catalog number: 51033597

IR Sensor

Stainless-steel interior

90-day warranty

Thermo Heracell VIOS 250i CO2 Incubator Features

Thermo Heracell CO₂incubators with non-corrosive stainless-steel interior chambers provide the ideal in-vitro environment. Large 255L capacity provides space for high throughput applications and accommodates spaces for culture devices for larger sample output.

Direct heat chamber

255 L (9.0 cu.ft.) electropolished stainless steel or copper chamber with compact footprint, easily stackable

Standard left hand door swing, field reversible by qualified service support

THRIVE active airflow

In-chamber fan gently distributes clean, humidified air throughout the chamber ensuring homogeneous conditions

Fast recovery of all parameters in 10 minutes or less for stable culturing conditions

Auto-stop function disables fan operation during door openings to minimize air exchange

ISO Class 5 HEPA filtration

In-chamber HEPA continuously filters the entire chamber air volume every 60 seconds

ISO Class 5 clean room air quality is achieved in the culture chamber within 5 minutes

Continuous protection minimizes airborne contaminants settling on interior surfaces

Covered Integrated humidity reservoir

Directly heated water reservoir provides stable high relative humidity levels, preventing culture desiccation

Cover limits particles and spilled media from settling into the reservoir, keeping water clean

Separating water provides a condensation free incubation chamber, limiting areas for contaminants to breed

5X faster humidity recovery than traditional water pan designs

Easy water handling: humidity reservoir may be filled or drained without removing shelves or cultures

Water level sensor indicates when a refill is needed

Steri-Run on-demand high temperature sterilization cycle

Independently validated overnight cycle ensures complete elimination of all biological contaminants

180°C cycle designed for uniform 12-log sterilization on all chamber surfaces

Simplifies cleaning and eliminates the need for separate autoclaving and reassembly of components

Intuitive, interactive iCAN touchscreen control

Bright LED screen to monitor all incubator interactions

Provides visibility to changes in culture environment with performance trend graphing, error and usage logs

Convenient on-screen user prompts and reminders for easy navigation

Selectable languages simplify operation: English, Spanish, German, French, Italian, Japanese and Mandarin

Wireless Monitoring Solution

Thermo Scientific DeviceLink HUB, our easy to install solution, works with the Thermo Fisher Cloud to monitor conditions inside your CO2 incubators.

DeviceLink HUB monitors chamber temperature, gas levels, alarms, and alerts from the incubator data output connection, RS-232 or USB

Alarms and notifications are also sent to the user by email and mobile push notification

DeviceLink HUB is an optional accessory for models with the iCAN touchscreen interface

DeviceLink HUB is available in the following regions

United States and Canada

28 members of the European Union

Iceland, Liechtenstein, Norway, and Switzerland

In situ sensors and probes designed for precise control

Dual temperature probes for over temperature protection with operational back up

Thermal conductivity (TC) CO2 sensor with rH compensation designed for fast recovery and stable operation

Testing

Most of our equipment is used and reconditioned. Our technicians test incubators using a calibrated digital chart recorder to verify the accuracy of temperature controls and readouts. We replace any hardware or parts as is necessary and clean all incubators thoroughly prior to shipment.

Pictures

The photos above are representative. Please bear in mind that the equipment shown is currently in storage in our warehouse and that in addition to testing each machine we will clean all the equipment thoroughly prior to shipment.

We often have multiple units of each model in stock and sometimes we're unable to feature photos of every single unit. At your request, we can provide photos of the equipment in your order after testing and cleaning.