Thermo Heracell VIOS 250i Copper-Lined Incubator

2026-03-02T05:05:55-05:00

Vendor: Thermo Scientific
Type: Incubators
Price: 10000.00

Thermo Heracell VIOS 250i Copper-Lined Incubator

The Thermo VIOS 250i Copper-Lined Incubator is specifically designed to deliver the performance reliability, ease of operation, and value required to support a range of culturing needs from basic research to the demanding, leading-edge applications, so you're ready for whatever comes next.

Thermo Heracell VIOS 250i Copper-Lined CO₂ Incubator Features

9.0 cubic foot capacity per chamber (255L)

TC CO2 sensor

iCAN touchscreen control

90-day warranty

Better solutions for optimal cell growth. Thermo VIOS 250i CO₂ incubators with pure copper interior chambers provide the ideal in-vitro environment. They feature THRIVE Active Airflow, where an in-chamber fan distributes clean, humidified air throughout the chamber ensuring homogeneous conditions; fast recovery of all parameters in ten minutes or less for stable culturing conditions; and an auto-stop function that disables fan operation during door openings to minimize air exchange.

The VIOS CO2 incubators also have a protected humidity reservoir where directly heated water provides stable high relative humidity levels, preventing culture desiccation; a condensation-free inner chamber that prevents a breeding ground for contaminant; and 5X faster humidity recovery than traditional water pan designs.

The direct heat chamber has adjustable, perforated shelving, and a reversible exterior door for added flexibility. The in-chamber HEPA continuously filters the entire chamber air volume every 60 seconds, and ISO 5 clean room air quality is achieved in the culture chamber within five minutes.

The iCAN touchscreen control is a bright, door-mounted display for easy accessibility and viewing. You can select between operation in English, Spanish, German, French, Italian, Japanese, and Mandarin.

Thermo Heracell VIOS 160i Copper-Lined CO₂ Incubator – specifications

Capacity (English) | 9.0 cu. ft.
Data Outputs | USB, 4-20 mA
Description | Single 255 L incubator including IR sensor, 4-20 mA data output, and factory compliance acceptance Pk.
Capacity (Metric) | 255 L
Chamber Material | Electropolished Stainless Steel
CO2 Concentration Range | 1 to 20%
CO2 Sensor Technology | IR Sensor
Lighting | LED
Material | Stainless Steel
Relative Humidity | >93% at 37°C
Temperature Range (Metric) | Ambient +3° to 55°C
Type | CO2 Incubator
Humidity Source | Covered integrated water reservoir
Dimensions (D x W x H) Interior | 24.8 x 23.9 x 26.4 in. (62.9 x 60.7 x 67 cm)
Dimensions (L x W x H) | 36.8 x 30.5 x 38.1 in. (93,4 x 77.4 x 96.8 cm)
Weight (English) | 215 lb.
Electrical Requirements | 120 V, 50/60 Hz

Testing

Our technicians test CO₂ incubators using a calibrated digital chart recorder to verify the accuracy of temperature controls and readouts, as well as using a Fyrite indicator to measure CO2 levels in the chamber. We replace any hardware or parts as is necessary and clean all incubators thoroughly prior to shipment.

Pictures

The photos above are representative. Please bear in mind that the equipment shown is currently in storage in our warehouse and that in addition to testing each machine we will clean all the equipment thoroughly prior to shipment.

We often have multiple units of each model in stock and sometimes we're unable to feature photos of every single unit. At your request, we can provide photos of the equipment in your order after it has been tested and cleaned.

Thermo HeraCell VIOS 250i CO2 incubator frequently asked questions

What’s the difference between copper-lining and stainless steel lining in the Thermo VIOS incubators?

Copper-lining offers an antimicrobial, self-disinfecting surface to reduce reliance on chemical disinfectants and are easy to maintain, ensuring a sterile environment. It can kill bacteria, fungi, and mold on surfaces to prevent contamination in cell cultures. Copper does tend to stain, but the patina and tarnish actually enhance the effectiveness of self-disinfecting.

Stainless steel units tend to be less expensive to purchase new. Stainless steel also requires regular cleaning and disinfection to prevent contamination. It is very durable though, and can withstand harsh cleaners. As the name implies it also resists staining and stays shiny and new looking for an extended period of time.

And what's the difference between the VIOS160i and 250i units?

Chamber size and footprint. The Thermo VIOS 160i chambers each have 5.8 cubic feet capacity whereas the VIO205i incubators have 9 cubic feet. The 250i also has larger exterior dimensions at 34.6”W x 25.1”D x 35.4”H.

I need multiple Thermo VIOS incubators for my lab. Are you able to supply several matching chambers or stacks?

Yes, we frequently outfit new labs with multiple pieces of equipment. We encourage you to check out some of the lab equipment start up packages from American Instrument Exchange. These include not only incubators but also biosafety cabinets, centrifuges, microscopes, and more. We offer easy customization to match your specific budget, brand preferences, and research requirements.

Do you offer repair services for the malfunctioning VIOS 250i incubators?

Absolutely. Our technicians can repair incubators and we offer both field service and depot repair services, depending on the nature of the problem and your location.